Shunt Calibration of a Strain Gauge Sensor
Uploaded by: Elena
On: 28 Jun, 2019
Viewed:106 times - 10 hour, 41 minute, 28 second ago
Downloaded: 0 times -
Definition - Shunt calibration is the known, electrical unbalancing of a strain gage bridge, by means of a fixed resistor that is placed, or “shunted”, across a leg of the bridge. The Wheatstone Bridge is widely used i...
Definition - Shunt calibration is the known, electrical unbalancing of a strain gage bridge, by means of a fixed resistor that is placed, or “shunted”, across a leg of the bridge. The Wheatstone Bridge is widely used in sensors such as load cells and torque sensors where shunt calibration is industry accepted method of in field calibration of strain gaged based sensors and instrumentation.
Purpose - Shunt calibration is a method of periodically checking the gain or span of a signal conditioner, which is used in conjunction with a strain gage based transducer, without exposing the transducer to known, traceable, physical input values. If required, adjustments can then be made to the signal conditioner to insure accurate measurement results.
Discussion - A strain gage bridge is “in balance” when the host mechanical structure is unloaded and unstressed. As the host structure (diaphragm, bending beam, shear beam, column, etc.) is loaded or stressed, the Wheatstone Bridge becomes unbalanced, resulting in an output signal that is proportional to the applied load. Shunt calibration simulates the mechanical input to a transducer by unbalancing the bridge with a fixed resistor placed across, or in parallel with, one leg of the bridge. For tension shunt calibration, the shunt resistor (R) is shunted across the [+] excitation (A) and [+] signal (B) leg of the bridge. For compression shunt calibration, the shunt resistor (R) is shunted across the [ –] excitation (D) and [+]signal (B) leg of the bridge
Method - The typical method of collecting shunt calibration data is as follows:
Connect the transducer to an appropriate strain gage signal conditioner and allow adequate time for the system to stabilize.
Apply a full-scale, N.I.S.T. traceable, mechanical input (or load) to the transducer.
Adjust the signal conditioner's gain or span controls, as required, to obtain a full-scale electrical output signal, and/or numeric display that represents the applied, mechanical input quantity.
Remove the mechanical input (or load).
Place a shunt calibration resistor across an appropriate leg of the Wheatstone Bridge as discussed above.
Record the value of the signal conditioner's output signal and/or numeric display. This value is the shunt calibration value, or equivalent load.
It is important to note that the shunt calibration value is specific for the particular shunt resistor used. This value, and the particular resistor, are now matched to the transducer and form the basis of the transferable shunt
The following formula can be used to estimate the approximate value of shunt resistor required to simulate a mechanical load
Rcal = (25000 * Rb) / (Output FS * Lcal)
Rcal = Shunt Resistor (ohms)
Rb = Bridge Resistance (ohms)
Output FS = Full Scale output of the load cell (mV/V)
Lcal = Load to be simulated, % of Load Cell Capacity
Summary - Shunt calibration is accepted throughout the industry as means of periodic calibration of a signal conditioner and transducer between calibrations of known, applied, traceable, mechanical, input values. Consequently, most all strain gage transducer manufacturers collect and supply shunt calibration data, along with a shunt calibration resistor, as a standard feature
Tags related to Shunt Calibration of a Strain Gauge Sensor - strain gauge shunt calibration calculator #current shunt calibration #strain gauge calibration curve #shunting a wheatstone bridge #strain gauge calculation example #calibration of strain gauge load cell #how to use a strain gauge #strain gauge formula
Most resistance sensor bridges are supplied by a grounded voltage or curreht source. Therefore the amplifier at the bridge's output should not have any of its input terminals grounded. In addition ...
Equation (3.31) shows that the sensitivity of a sensor bridge depends on the supply voltage V, on the sensor's resistance under no load condit-:ns Rs,
and on the ratio k between the resistance...
Resistance temperature detectors (RTDs) are commonly used in industrial and scien-tific temperature measurements. The most common types are pure platinum (Pt) formed into wire or evaporated in a th...
Wheatstone bridge circuits have been in the field for a very long time and still are among the first choices for front-end sensors. Whether the bridges are symmetric or asymmetric, balanced or unba...
To obtain a voltage proportional to any size changes in one ofthe resistances in a Wheatstone bridge, we can either modify the structure of the bridge or
perform analog processing of the outpu...
A common way of obtaining an electric signal in a Wheatstone-bridge-based measurement is with the deflection method. Instead of measuring the action
needed to restore balance in the bridge, th...
As the fastest growing demand of circuit and wiring diagram for automotive and electronics on internet based on different uses such as electronic hobbyists, students, technicians and engineers than we decided to provide free circuit and wiring diagram base on your needed.
To find circuit and wiring diagram now a day its easy. E-learning through internet as a right place to search an exact circuit and wiring diagram of your choice and it's much fun and knowledgable. On internet you will find thousands of electronic circuit diagrams some are very good designed and some are not. So you have to modify them to make them according to your needs but some circuits are ready to make and require no changes.
There are many categories of circuit and wiring diagrams like automotive, audio circuits, radio & RF circuits, power supply circuits, light circuits, telephone circuits, timer circuits, battery charger circuits etc. There are many types of circuit and wiring diagrams some are very easy to build and some are very complicated, some are so small and some contain huge list of parts.
We provides free best quality and good designed schematic diagrams our diagrams are free to use for all electronic hobbyists, students, technicians and engineers. We also provides a full educational system to students new to electronics. If you are new to electronics you are a student or a electronic hobbyist and want to increase your knowledge in electronics or want to understand electronics in a very easy way so this is the right place for you we provide electronics beginner guide tutorials to easily understand complicated electronic theory. Our mission is to help students and professionals in their field.